Size: 4.05" (10.3 cm) H x 5.65" (14.4 cm) W x 1.95" (4.9 cm) D
Weight: 14.4 ounces (408 g)
Flow Rate: 0.1 mL/hr from 0.1 - 10 mL/hr; 1 mL/hr from 10 - 600 mL/hr
Flow Rate Accuracy: +/- 5% of selected flow rate
Dose Limit: 0.1 mL from 0.1 - 10 mL; 1 ml from 10 - 3000 mL
Operating Orientation: Any
Audio/Visual Alarms: Load Set(Set not load correctly); Low Battery, No Flow In (upstream occlusion), No Flow Out (downstream occlusion), No Food (Air in Line/Empty Container), Push Run to Feed (Hold/Pause over 2 minutes), Shut Door(Door open while running), Chek (Rate,Dose and Interval incompatible), Erra Erraz (Self Check Failure)
Memory: Settings retained until use changes
Battery: Lithium Ion; 24 hours at 125 mL/hour; 6 hour recharge time
Maintenance: No annual maintenance or calibration required
Cleaning: Washable under running water (water resistant)
Health Canada Licence No 65642 Device Identifier INFKIT2
EnteraLite® and Infinity® are trademarks of MOOG, Inc.
FOR COMPLETE INDICATIONS, CAUTIONS AND INSTRUCTIONS
FOR USE, SEE LABELLING SUPPLIED WITH DEVICE.
Moog is a registered trademark of Moog, Inc. EnteraLite® and Infinity® ∞ logo® are registered trademarks of Zevex, Inc.